How to Create a Digital Pulse Train with Output Voltage Greater than 5V
Hardware: CompactDAQ>>Digital Output Modules>>NI 9472, CompactDAQ>>Digital Output Modules>>NI 9474
I would like to create a digital pulse train with an output voltage value greater than 5V. Is this possible?
Yes. This can be achieved with a hardware timed sourcing digital output module. For instance, the 9472/4 is a hardware timed module which can take an external supply voltage above 20V. If you tie the external supply to VSUP and use a counter to generate a pulse train you will achieve a pulse train at the voltage of the supply.
KnowledgeBase 48FAAMCW: Troubleshooting Digital Output and Pulse Generation on the NI 9472, 9474, 9476 & 9475.
Report Date: 02/23/2011
Last Updated: 04/18/2011
Document ID: 5IM77EKU