Archived: Using cDAQ with DAQ Signal AccessoryThis document has been archived and is no longer updated by National Instruments
Primary Software Version: 1.0
Primary Software Fixed Version: N/A
Secondary Software: LabVIEW Development Systems>>LabVIEW Base Development System
I'm trying to use the DAQ & Signal Conditioning course with a cDAQ setup. The instructions reference connecting the cDAQ to a DAQ Signal Accessory. A DAQ Signal Accessory is powered by a 68-pin cable and many of the signals referenced are internally wired and are only connected through the 68-pin female 0.050 series D-type connector, how can I use the DAQ Signal Accessory with this course?
To use a cDAQ system with the DAQ signal Accessory it is necessary to use the DAQ Signal Accessory Daughter Card to power the Signal Accessory and break out the internal signals. The daughter card breaks out the 68-pin female 0.050 series D-type connector on the signal accessory to gain access to the following pins that are not directly available to the user.
If you would like to purchase a DAQ Signal Accessory Daughter Card please contact your local field sales engineer and reference Part Number: 198821A-01. This product is currently unavailable for purchase on National Instruments' website.
It should be noted that if you do not have a DAQ Signal Accessory, it is possible to replicate its functionality with a few LEDs and a function generator.
Figure1: DAQ Signal Accessory with Daughter Card
Figure 2: DAQ Signal Accessory Daughter Card
Related Links: Contact NI
Report Date: 06/03/2009
Last Updated: 11/20/2015
Document ID: 4Y2BTHH7